A hospital grade, digital and automatic blood pressure monitor with patented SureBP™ technology (bluetooth optional). Supplied with 1x size 11 standard adult 25-34cm cuff and 1x size 12 large adult 32-43cm cuff.
Small hand held form factor
Easy to carry and record readings in triage settings or rounds
SureBP™ NIBP Technology, enabling accurate and fast 15 second blood pressure readings
USB-wired and Bluetooth enables connectivity to select EH/EMR systems, improving accuracy and efficiency with documentation
Compatible with the FlexiPort cuff system, available in sizes ranging from small child (size 08) to thigh (size 13)
A single charge can take approximately 100 readings due to Lithium-ion battery technology
3 year warranty applies to Connex ProBP 3400 Digital Blood Pressure Device (1 year warranty on battery)
Supplied with 1x size 11 standard adult 25-34cm cuff and 1x size 12 large adult 32-43cm cuff
Blood pressue accuracy meets or exceeds ANSI.AAMI SP10:2002.
Cuff pressure range: 0 to 300 mmHg
Systolic range: 60 to 250 mmHg
Diastolic range: 30 to 160 mmHg
Mean Arterial Pressure (MAP) range: 40 to 190 mmHg
Pulse rate range: 35 to 199 bpm
Pulse rate accuracy: ±5%
Overpressure cut off: 300 mmHg ±15%
Dimensions: (H*W*D) 15 cm*8 cm*5.6 cm
Weight: 450 gramms
Power requirements: 100 - 240 VAC, 0.18A, 50-60Hz
Battery: Lithium-ion, 3.7V, 210mAh, 7.8Wh
FlexiPort blood pressure cuffs, reusable, ranging from small child (Size 08) to thigh (Size 13)
Very pleased. It takes blood pressure very quickly. I find if the patient is talking, or otherwise making the reading difficult, it needs to take the BP the traditional way (on deflation rather than inflation). Hardly ever fails to take a reading, unlike my old machine which produced an error about 50% of the time. One down side is that the screen turns off automatically after just a minute or two, and I don't know if you can set it to stay on. And the other minor criticism is that you need to go and find a screwdriver to assemble the machine out of the box.
I am not a medical professional. I purchased this unit for home use because I was after a calibrated automatic blood pressure monitor that would provide more consistent readings than my old unit.
The 3400 ProBP is compact, easy to use, provides consistent results and is adequate for my needs. There are a range of cuff sizes available and the battery can be replaced. My only complaint is that the cuff can be a bit tricky to fit with one hand when taking your own blood pressure.
The purchase price from AMA Medical products was the best I found but the experience with delivery was very poor. Why won't you deliver to a PO Box? I can't afford to sit around all day waiting for a courier who may or may not arrive on the day. I had two missed delivery attempts and an incomplete notification resulting in me not knowing where the package was and having to waste my time trying to track it down. Again, why won't you deliver to a post office? If you would then none of this would have happened.
I had an Omron which spat the ***** every time it hit an ectopic. The Welch Allyn is a big improvement, faster, no Err readings and easy to change cuff size. My only whinge is that I would have liked a bar cuff. Buy it and you won’t be disappointed. I will pay for itself in time saved never mind the improvement in your BP.